Fitness trackers make it possible to exercise smarter. These wearable devices gather information on sleeping patterns, eating habits and training sessions, before presenting them in graphs that are easy to understand. Owners can use this information to make decisions about what they eat, the kind of exercises they do and their workout intensity.
A heart rate monitor is becoming more common on fitness bands with this feature available on more than half the models we tested. Wearing a stand-alone heart rate monitor is difficult because of their large size, but by integrating them with an unobtrusive fitness tracker, it should be easier for people exercising to identify if they're in a heart rate zone ideal for burning fat, improving endurance or increasing performance.
While we discovered some of the built-in heart rate monitors may not be perfect, our latest fitness tracker review revealed small improvements across the market. In this article we reveal what we found when we tested over 60 popular fitness trackers with and without heart rate monitors from brands such as Fitbit, Sony, Nokia, Garmin and Apple.
Wasting a workout
A heart rate monitor showing a reading that is even marginally off is enough to undermine the efforts of a workout, says Dr Kate Edwards, the course
director of Exercise Sports Science at the University of Sydney.
People exercising should aim for a specific heart rate zone to reap the results they're after. The light zone, for instance, is the ideal zone for fat
burning, while the moderate zone targets aerobic fitness.
But a difference of five percent of your maximum heart rate is all it'd take to diminish the results of a workout. "The zone is a ten percent range,"
says Edwards. If you were exercising at 75% of your maximum heart rate and your monitor said you were at 81%, you would be working in a different zone.
"As soon as those intensities aren't accurate, then the benefits aren't there."
Not getting the benefits could affect your motivation to exercise, says Edwards, as well as your ability to achieve your goal.
It's easy to see how inaccurate readings can short-change the effort poured into dieting, sleeping and exercising.
How we test
We tested the heart rate monitors of over 40 fitness trackers at the Faculty of Health Sciences at the University of Sydney. A 12-lead electrocardiography (ECG)
monitor mapped the real-time heart rate of a professional athlete as he ran on a treadmill in the faculty's Sports Chamber. The readings from the ECG
monitor – a device Dr Edwards described as the most accurate heart rate monitor, capable of identifying an immediate change in heart rate – were compared to
the readings of each of the fitness trackers.
Readings were taken at one-minute intervals for five minutes to compare the accuracy of each tracker to that of the ECG. To compare how well the trackers
respond to sudden changes in intensity – typical to interval training – we also took readings at 10-second intervals in the first minute of warming up,
and then the sixth minute when cooling down.
Testing heart rate monitor accuracy at the University of Sydney in 2016.
Performing the same heart rate monitor test in 2017.
Of the trackers equipped with a heart rate monitor, 23 fell within an acceptable margin of error when measuring active heart rate. Among the best
performing were the monitors in bands from Garmin, Fitbit, Under Armour and Striiv, as these managed to produce readings within
95% accuracy – despite having to contend with motion and sweat. The Samsung Gear S3 Frontier smartwatch was the best performing with an overall accuracy of 100%, but it can only read and display data when your arm stops moving. However, the Apple Watch Series 2, Samsung Galaxy Watch 4, Mio Alpha 2 and TomTom Touch return 99% accuracy even with movement, which makes them a much more effective training option.
Resting heart rate results fared better. We assessed these against industry-grade GE Healthcare Carescape V100, Dinamap V100 and Dinamap V150 heart rate monitors since first testing fitness bands in 2014. All models are calibrated and set to record data under the same conditions. The only difference is in input and export tools (the V150 is the first model to introduce a touch screen for example). Most trackers fell within the acceptable margin of error.
Testing resting heart rate accuracy against the GE Dinamap V150.
Pointless fitness trackers
Pausing in the middle of a workout to read a heart rate can undermine its intensity, but that's exactly what the fitness trackers from Apple and Samsung
demand. These fitness trackers can't provide a reading in real time, so at key points in a run or a bike ride, you'll have to stop, hold your hand level to
the ground, make sure it's quiet and wait between 20 to 30 seconds for your reading. But by then the reading is irrelevant, as a heart rate can drop by 20
beats during the period of inactivity.
Edwards believes the manufacturers have adopted this method because it may be more accurate. However, the need to stop moving and the time needed to take a reading meant we couldn't test how well these trackers respond to
rapid changes in intensity, such as during the warming up and cooling down periods of a workout. For this reason, the active heart rate results for all trackers in our test are
based on the recordings taken at minute intervals, and don't take into account the readings from the warm-up and cool down.
Waking heart rate readings
The heart rate monitors in most fitness trackers performed better in our resting heart rate test. Capturing readings every morning can help identify any
changes in overall health; a spike in your waking heart rate can reveal injury, illness and even long-term health problems.
This part of the test took place in the CHOICE labs, using the GE Carescape V100 heart rate monitor to measure against. Most of the fitness trackers we
tested captured resting heart rate readings within 95% accuracy, as sitting steadily, waiting for 30 seconds and not having to contend with sweat helped them hone in on an accurate reading.
But sitting and waiting never helped anyone lose weight or gain strength. Edwards recommends using a chest strap heart rate monitor when exercising,
although there are other methods that can be used to deduce the intensity of a workout. The rate of perceived exertion, a scale judging the ease or
difficulty of an exercise, is one.
"But there are simple ways, like the talk test," adds Edwards. "If you can't comfortably talk [while running], then it's a higher intensity."
In other words, take a running companion – of the human variety.